The Xinzhuang (LNB) brand developed by Shanghai Xinzhuang Instrument Co., Ltd. is widely used in microbial tissue cell culture, seed germination, seedling experiment, plant culture, etc. It can simulate different environmental and climatic conditions.
For the maintenance and cultivation of the lighting incubator in Shanghai Xinzhuang, please refer to the following information:
1. The incubator housing should be reliably grounded.
2. The incubator should be placed in a cool, dry, well-ventilated place away from heat and sunlight. Place it smoothly to prevent noise from occurring.
3. In order to ensure the effective heat dissipation of the condenser, the distance between the condenser and the wall should be greater than 100mm. There should be a 50mm gap on the side of the box and at least 300mm on the top of the box.
4. When the incubator is transported, repaired and maintained, it should avoid collision, shaking and shaking; the maximum inclination is less than 45 degrees.
5. The instrument suddenly does not work. Please check whether the fuse tube (after the box) is burnt out and check the power supply.
6. When the incubator is working in refrigeration, it is not advisable to make the difference between the temperature inside the box and the ambient temperature greater than 25 degrees.
1. The compressor is not cooled, the temperature inside the incubator is not uniform, or the circulating air outlet is frozen.
2. In the balanced mode, the compressor is always in working condition, the cooling effect is better and the compressor can be effectively maintained.
3. If the temperature is reduced to a certain value when working in a single system, the temperature will fluctuate greatly when the door is opened or closed. Causing the compressor to be ordinary
4, the start is easy to shorten its service life.
5, the temperature is not uniform:
6. It may be that the discharged items are too dense or the circulating fan inside the box does not work.
7. If you use long light, the light bulb will easily heat up and the temperature will be uneven.
The outlet is frozen:
Each refrigerated instrument has a defrost cycle during which the compressor does not cool. The ice on the condenser is melted by the heating element. The melted water is likely to collect at the air outlet and eventually cause ice to form.
Shanghai Xinzhuang Instrument Co., Ltd. specializes in the production of intelligent light incubators, biochemical incubators, constant temperature and humidity incubators and other laboratory equipment. For more information, please continue to pay attention to Shanghai Xinzhuang, Shanghai Xinzhuang welcome your inquiry!
Our Biodegradable Disposable Paper Cup is only for one-time use, effectively preventing cross-infection of infectious diseases.All Disposable Biodegradable Tableware have been tested by American ASTM-D6400 and EU EN13432 compostable certification standards.The used biodegradable disposable paper cup is buried in the soil, or buried in a micro-organism environment. After 3 months of composting, the paper cups can be completely degraded into carbon dioxide and water 100%, becoming organic fertilizers without white pollution.
PLA Disposable Paper Cups
Ningbo ECO Biodegradable Technology Co., Ltd , https://www.chinabiodegradable.com